The tissue hook and electrodes of the Radnoti Electrode Holder are small enough to fit into the standard 25ml chamber that is included with the ML1110.
However, the arm of the holder is not long enough to completely reach the chamber if held in the upper part of the stand of the ML1110. It reaches sufficiently far enough for the electrodes and tissue hook to enter the organ bath chamber, but only at an angle, rather than a completely vertical orientation, and is therefore not ideal. The best solutions for this issue are:
- Use the relevant PanLab stimulating electrodes and apparatus.
- Use an additional clamp stand to hold and position the Radnoti Holder nearer the chamber.