For General Cleaning:
The flowprobe sensor tips can be cleaned by using 70% ethanol solution. The probe tips can be left to soak overnight if extended cleaning is required. The cabling / connector should not be put in contact with the ethanol.
The flowprobes can be autoclaved if decontamination / sterilization is required, and have been tested up to 20 autoclaving cycles without issue. The entire probe is suitable for autoclaving, including the connector and cabling.