Trigno Wireless Device Enabler 1.3.4 Windows
Trigno Wireless Device Enabler v1.3.4 is compatible with LabChart v8.1.10 or later and Trigno Control Utility 3.5.7.
Base Station and Sensor firmware should also be updated to the latest version using the Delsys Software Update Utility.
Notes for this release
- Trigno Wireless 1.3.4 adds support for the new Delsys Quattro and Delsys Galileo sensors types.
- The Quattro and Galileo sensors must be paired to a slot in the top row of the Trigno Control Utility with no other sensors paired to another slot in the same column.
- Trigno Wireless 1.3.4 device enabler supports Trigno Server 3.5.7
- The Trigno Wireless 1.3.4 installer must be acquired from either the adinstruments.com website or the CD along with Trigno Server 3.5.7, it is not available for download from LabChart Feature Manager
- Trigno Wireless 1.3.4 is not compatible with any versions of Trigno Server prior to 3.5.7 or any versions after 3.5.7
- The Trigno Control Utility may require sensors to be updated to the latest firmware. Please use the Delsys Software Update Tool v22.214.171.124 for legacy sensors or the Delsys Firmware Update App on Android OS for bluetooth sensors.
- Pairing the Quattro or Galileo to a slot not in the top row
- Avanti type sensor modes can’t be changed.
- EMG sensor bandwidths can’t be set via LabChart in this version of Trigno Wireless, the default is the Wide bandwidth mode.
- Avanti X and Y Magnetometer axis are inconsistent with previous IMU sensor Magnetometer axis.
- Changing the Powerlab gain/sampling settings whilst sampling alongside a Trigno sensor can cause synchronization issues. Please change these settings when you are not sampling.
- By default, LabChart will load the sensors in the same order they are paired in the Trigno Control Utility. When opening settings files, please check that you have the same order, otherwise there may be a channel naming to sensor input mismatch.
- When changing sensor configurations in between LabChart sessions, make sure all undesired sensors are turned off (use the ‘Sensors Off’ button) and relaunch the Trigno Control Utility. Even this can sometimes result in a ghost sensor in the next session.
- The XY orientation with respect to the sensor’s arrow indicator is not consistent across sensor types, positive X and Y axis are swapped for the legacy EMG type A sensor and newer EMG type sensors.
- The Load cell sensors need to be calibrated using LabChart’s Units Conversion feature to obtain accurate readings based on specific load cells.
- Sensors may have offsets and any offsets can be compensated for using an arithmetic expression in LabChart.
- Changes to the Trigno Control Utility’s settings are not supported by LabChart, this includes changing the orientation filter for the IMU sensor, changing sensor modes and settings from within the Control Utility and changing SDK communication port numbers. The modes of the sensor should be set using LabChart’s range dropdown setting.
- Trigno Server 3.5.7 is not supported on 32-bit operating systems