Glass microelectrodes are used for extra- or intracellular recordings. These have usually a high resistance of 5 to 10 G?.
ADInstruments' Bio Amps have an input impedance of 200 M? differential and therefore can only be used with low impedance metal electrodes. Glass electrodes require a high impedance differential amplifier, usually in combination with an active head stage.