DO NOT use cleaning agents containing iodide. Always rinse the transducer in water after the use of cleaning agents. Make sure the transducer is clean and dry before storing.
DO NOT autoclave or use an ultrasonic process on the transducer. Max temp 70 degrees C.
DO NOT immerse connector into any kind of liquid.